Manometers MP KIMO MP KIMO MP Data Sheet ( KB) For detailed information please refer to: [email protected] · Environnement SA. This is the MP Thermo-Anemo-Manometer. It is an easy to use meter which can take complex measurements. The MP is designed to measure Air. KIMO Instruments portable thermo-anemo-manometer with backlit digital display. Functions: pressure, temperature, air velocity and airflow with Pitot tube, CO.
|Published (Last):||11 October 2004|
|PDF File Size:||5.68 Mb|
|ePub File Size:||13.83 Mb|
|Price:||Free* [*Free Regsitration Required]|
In regards of Quality insurance norms, we recommend that the instruments are checked once a year.
Select Start for dataset launching. Use the arrows to modifiy the threshold value then press Validate to validate alarm threshold and back to measurement screen. When the countdown is finished, the CO max is displayed.
Kimo MP H MPH Thermo-Anemo-Manometer
Select other dataset using arrow keys t and u. Create or launch a continuous dataset Select dataset A continuous dataset can be carried out using MP and is composed of several dated measuring points. In regards of Quality insurance norms, we recommend that the instruments are checked once a year. Battery When battery indicator flashes it is recommended to change the batteries: When CO peak is selected, the period is diplayed 30 seconds by default.
Kimo MP200 Thermo-Anemo-Manometer
Enter text from picture: Memory capacity of the instrument is up to 8, points or 50 datasets. The number of points selected and the parameter are displayed.
Select Start with access key for launching measurement. You can access this function selecting COmax with the access key CO max The CO is measured on an adjustable period, the maximum value measured in this period is klmo CO max. It’s a community-based project which helps to repair anything.
When the solenoid valve is enabled, it runs every minute. Remove the front part at the back of the instrument.
Comments to this Manuals Your Name. From this screen you can: You can choose between a planned or a continuous dataset. Average – Rec Recording Veloci.
Memory capacity of the instrument is up to 8, points or 50 datasets. Launch a planned dataset A planned dataset is composed of several locations.
MP – KIMO – PDF Catalogs | Technical Documentation | Brochure
You must enter duration: Once pressing Autoz key, the zero is readjusted. From this screen, you can: It is thus necessary to enter the real temperature of use to obtain more precise results. To modify the period, press Period with the access key. This function can also be automatically performed by using the solenoid valve function.
Summary screen of selected dataset is displayed From this screen, you can: Don’t have an account? Air velocity with Pitot tube: By pressing ‘print’ button you will print only current page. Go to Recording menu. Click on Mesure to display values kkmo of selected dataset.
Kimo MP 200 P User Manual
On enabled mode, the probe is connected, the measurement is carried out and the value is displayed. This menu allows to view probe Probe display information plugged to C2, Module, C1 or wireless connections.
To save your dataset click on Save with the access key. Go to ,imo menu. Press 1x in order to select HOLD function: You can view each measuring points if you click on Visu. Preview of tables of points of datasets Dataset summary You can display tables of points of datasets performed on the device.