The Synergy HT Multi-detection reader is a robotic-compatible microplate .. and J. Sambrook () Molecular Cloning A Laboratory Manual, Cold Spring. View and Download BioTek Synergy HT operator’s manual online. Multi-Mode Microplate Reader. Synergy HT Laboratory Equipment pdf manual download. Manuals and User Guides for BioTek Synergy HT. We have 2 BioTek Synergy HT manuals available for free PDF download: Operator’s Manual.
|Published (Last):||11 February 2015|
|PDF File Size:||5.65 Mb|
|ePub File Size:||1.58 Mb|
|Price:||Free* [*Free Regsitration Required]|
BIOTEK SYNERGY HT OPERATOR’S MANUAL Pdf Download.
Remove The Shipping Panel Do not block any air vents. Removing the microplate carrier shipping screw Replace the microplate carrier shipping Important: Page 45 Hold the syringe from rotating while tightening the thumbscrew.
Reference Field Change Notice L Enter text from picture: Waste Electrical And Electronic Equipment Tip Prime before dispensing. Introduction Chapter 1 Introduction This chapter introduces the Synergy HT, describes its key features, lists its package contents, and provides contact information for technical assistance.
Running A Dispense Protocol optional Chapter 5: Instrument Dimensions For Robotic Interface The TR cartridge allows light from the xenon flash bulb to be input to the fluorescence optical system within the Synergy instrument.
Page PCB sent the wrong voltage request to the lamp power supply. Updated Appendix C, Error Codes. Technical Assistance Center tac Contact BioTek Customer Care with any questions.
Creating The Test Protocols Chapter 4: Stream water from a faucet through the pipe to be sure it is clean. Creating Protocols And Experiments Return the jig to BioTek for a replacement jig.
Biotek Synergy HT Manuals
Insert the top foam end cap. Page see instructions on the next page. Since the standard deviation is less than this value, the reader meets the test criteria.
Initial setup of the dispense module BioTek Instruments, Inc. Turn to Chapter 5, Preventive Maintenance Do not block any amnual vents. Required Materials BioTek offers a liquid test kit PN containing the microplate and solutions used in this procedure.
Controlling The Dispense Module Performance Check After reassembling the instrument, perform the following to verify that the instrument biohek functioning properly: Page 74 Gen5 provides a special utility to purge fluid from the dispense tubing and syringe by pumping the fluid in reverse, back into the supply bottle. Saturation Errors filter wheels hominga error could indicate a missing filter.
Page 85 BioTek Instruments, Inc.
In Gen5, this option is found in a Read step within a Procedure. Page 17 Some mxnual of the dispense module can present pinch hazards when the instrument is operating.
The fluid should pump through the tubing and dispense into the priming plate. Page This test is performed when the instrument is first turned on and then tested periodically during background functions. Chapter 2 Installation This chapter includes instructions for unpacking and setting up the Synergy HT and, if applicable, the external dispense module.
Page This error indicates that the calibration data has been entered but the checksum failed.